Sample rate = 200kS/s (max)
Measurement range = 10V
Low Pass filters = OFF
Measured with 5000V/us pulse as a reference.
Typical Slew rate of all of the above amplifiers is 1V/us.
Typical measured result for rise time was 2 samples (t=5us @ 200kS/s) for the change from 0V to 10V so SR = 10V/(2*5us) = 1V/us.